Internal resistance determination EIS – Electrochemistry & Battery- Application Note 38Latest updated: May 6, 2020
In this application note we demonstrate how to determine the dynamic resistance on a LiFePO4 battery using a DC technique (GCPL5), and those results compared with that of an impedance spectroscopy technique (PEIS). The conclusion of this comparison is that the dynamic resistance can be used as a rough approximation of the impedance data determined by EIS and that a wide frequency range EIS is still best to fully characterize a battery.
Currently, several types of batteries are investigated (Li-ion, LiFePO4, ….). Each type has its own specifications such as nominal voltage (V), capacity (Ah), power (W), energy
(Wh), cyclability… The apparent resistance RΔt of the battery is also a relevant parameter. This resistance can be determined by a simple current interrupt technique. In this technique,
a current pulse is applied to the battery and the voltage shift is measured after a certain period of time Δt (Fig. 1). The ratio of the voltage shift ΔE and the current difference ΔI yields to a resistance, R:
To view the entire application note please click the download button below.